Open as PDF
Installation Instructions 3 - 1
3 INSTALLATION INSTRUCTIONS
This section contains instructions for the followin
the contents of the shipment for completeness and/or dama
transport (initial inspection)
the safety of the operator
the power supply
the input and output connections
the installation acceptance checks
3.2 INITIAL INSPECTION
Inspect the contents of the shipment for completeness and note whether any
e has occurred durin
transport. If the contents are not complete, or if there
e, inform your FLUKE Sales and Service Center so that repair or
replacement of the instrument can be arran
Pay special attention to the followin
• accessories to be supplied (refer to Appendix A).
• front panel keys and display.
• rear panel connectors and terminal blocks.
• cabinet surfaces.
The shipment must contain one power supply plus the accessories supplied. For
a complete list of accessories supplied, refer to the INITIAL INSPECTION pa
in the front of this manual.
3.3 OPERATOR SAFETY INSTRUCTIONS
The instrument has been desi
ned and tested in accordance with IEC Publication
1010 for Class 1 instruments and has been supplied in a safe condition. This
Operation Manual contains WARNING and CAUTION statements, that must be
followed by the user to ensure safe operation and to keep the instrument in a safe