Open as PDF
5 - 24 Users Manual
Step automatically throu
h the predefined list of volta
e and current
s stored at the memory addresses 20 to 40. When the end of the list
is reached, it must be started a
ain at the be
. The next volta
must be output for channel 1 at an interval time of 5
Press to enter the STEP submenu.
Press to set the AUTO STEP function on.
Press to set the TIME INTerval function.
Press to set the interval time at 5 seconds.
Press to set the CONTINUOUS step function.
Press to set the CONTINUOUS step function on.
Press to set the START AT function.
Press to set the start address of the sequence list at 20.
Press to set the STOP AT function.
Press to set the stop address of the sequence list at 40.
Press to leave the AUX menu.
Press (only for multiple output models) to select output channel 1.
Press to enable output channel 1.
Press to make the OPERATE mode active.
Press to enter the step mode.
Press to select start address 20 of the list.
The display reads : 1 S T E P T O 2 0
Press to initiate the automatic steppin
• Check that the memory address is automatically incremented every
5 seconds on the display.
• Exit the automatic steppin
the (backspace) key. The last
e and current settin