Open as PDF
5 - 28 Users Manual
GPIB control can only be done in the remote mode of operation. An external
er can be pro
rammed in the followin
• Send →
TRG By sendin
the common command
TRG to the power
• Send → GET By sendin
the GET (Group Execute Tri
er) code over
the GPIB interface.
5.6 ADVANCED USE
This section describes how to connect several output channels (with the same
s) to obtain hi
her output power throu
h a connected load.
To increase the total output current throu
h a connected load, the output channels
must be connected in parallel. Refer to section 5.6.1.
To increase the total output volta
e over a connected load, the output channels
must be connected in series. Refer to section 5.6.2.
This section also describes the way to connect more than one load to an output
channel (multiple loads). Refer to section 5.6.3.
CAUTION: Only output channels with equivalent volta
e and current ratin
be connected in series or in parallel to prevent that the channel with
the lowest output ratin
s will be dama